Potřebujeme váš souhlas k využití jednotlivých dat, aby se vám mimo jiné mohly ukazovat informace týkající se vašich zájmů. Souhlas udělíte kliknutím na tlačítko „OK“.
Standard Test Method for Measurement of Focal Spots of Industrial X-Ray Tubes by Pinhole ImagingPřeložit název
NORMA vydána dne 1.12.2020
Označení normy: ASTM E1165-20
Datum vydání normy: 1.12.2020
Kód zboží: NS-1029228
Počet stran: 17
Přibližná hmotnost: 51 g (0.11 liber)
Země: Americká technická norma
Kategorie: Technické normy ASTM
focal spot, pinhole camera, pinhole imaging, X-ray, X-ray tube,, ICS Number Code 19.100 (Non-destructive testing)
|Significance and Use
5.1?One of the factors affecting the quality of radiologic images is the geometric unsharpness. The degree of geometric unsharpness is dependent on the focal spot size of the radiation source, the distance between the source and the object to be radiographed, and the distance between the object to be radiographed and the detector (imaging plate, Digital Detector Array (DDA) or film). This test method allows the user to determine the effective focal size of the X-ray source. This result may then be used to establish source to object and object to detector distances appropriate for maintaining the desired degree of geometric unsharpness or maximum magnification for a given radiographic imaging application, or both. Some ASTM standards require this value for calculation of a required magnification, for example, Practices , , and .
1.1?The image quality and the resolution of X-ray images are especially sensitive to the characteristics of the focal spot. The imaging qualities of the focal spot are based on its two dimensional intensity distribution as seen from the detector plane.
1.2?This test method provides instructions for determining the effective size (dimensions) of standard and mini focal spots of industrial X-ray tubes for focal spot dimensions from 100 ?m up to several mm of X-ray sources up to 600 kV tube voltage. Smaller focal spots down to 50 ?m could be evaluated with less precision. This determination is based on the measurement of an image of a focal spot that has been radiographically recorded with a pinhole technique. An alternative method with a plaque hole IQI may be found in the Annex A, which covers the same focal spot sizes.
1.3?Smaller focal spots should be measured using Test Method using the projection of an edge.
1.4?This test method may also be used to determine the change in focal spot size that may have occurred due to tube age, tube overloading, and the like. This would entail the production of a focal spot radiograph (with the pinhole method) and an evaluation of the resultant image for pitting, cracking, and the like.
1.5?UnitsValues stated in SI units are to be regarded as the standard.
1.6?This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.7?This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|2. Referenced Documents
Chcete pravidelně odebírat informace o nově vycházejících normách z celého světa a to zcela zdarma?
Přihlašte se k odběru. Vše je velice jednoduché a absolutně ZDARMA.
Na výběr máte vydavatele z celého světa.
Poslední aktualizace: 29.02.2024 (Počet položek: 2 880 691)
© Copyright 2024 NORMSERVIS s.r.o.