Potřebujeme váš souhlas k využití jednotlivých dat, aby se vám mimo jiné mohly ukazovat informace týkající se vašich zájmů. Souhlas udělíte kliknutím na tlačítko „OK“.
Standard Test Methods for Measurement of the Rotational Viscosity of Paints, Inks and Related Liquid Materials as a Function of TemperaturePřeložit název
NORMA vydána dne 1.12.2020
Označení normy: ASTM D7867-13(2020)
Datum vydání normy: 1.12.2020
Kód zboží: NS-1013186
Počet stran: 3
Přibližná hmotnost: 9 g (0.02 liber)
Země: Americká technická norma
Kategorie: Technické normy ASTM
coaxial cylinder viscometer, cone/plate viscometer, rotational viscosity, viscosity, viscosity as a function of temperature,, ICS Number Code 87.040 (Paints and varnishes),87.080 (Inks. Printing inks)
|Significance and Use|
5.1The viscosity of paint, inks and many related liquid materials is dependent on temperature. It is useful to know the extent of this dependence. One use of such information is to prepare a viscosity-temperature table or curve. Then, if ambient conditions do not allow the measurement of viscosity at the exact temperature stated in a specification or regulation, the viscosity measured at ambient temperature can be used to determine the viscosity at the temperature of interest through the use of the previously prepared table or curve. Viscosity measurements that cover a range of shear rates as well as temperatures could include shear rates associated with paint application or allow extrapolation to such shear rates. This information would enable a producer or user to estimate the effect on application of heating the paint.
1.1These test methods cover the use of rotational viscometers to determine the dependence of apparent viscosity of paints, inks and related liquid materials on temperature. The first method uses a standard rotational viscometer with concentric cylinder geometry running at a fixed rotational speed as the temperature is increased or decreased. The second method uses a rotational viscometer with cone and plate geometry running at a fixed rotational speed as the temperature is increased or decreased. The third method uses concentric cylinder or cone/plate geometry operated with a shear rate ramp at several discrete temperatures.
1.2The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3This standard may involve hazardous materials, operations and equipment. This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.4This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|2. Referenced Documents|
Poslední aktualizace: 27.01.2023 (Počet položek: 2 662 456)
© Copyright 2023 NORMSERVIS s.r.o.